Levin, et al. (2019) describe the introduction of an electronic hand hygiene (HH) surveillance and intervention system into the general ICU of a tertiary care teaching hospital, the obstacles to success and reasons for the system’s ultimate failure and removal.
The system was based on radiofrequency transmitters in patient areas, on HH dispensers, and individual personal bracelets. The transmitters were connected to a central computer. The system was designed to detect entry and exit from patient areas and provide real time alerts of missed HH performance.
A staff satisfaction questionnaire was administered followed by validation of system accuracy. Electronic data were compared to human observer data collected during defined observation periods.
Data from 41 questionnaires revealed low satisfaction rate (21/41, 51%). Low system accuracy (31/41, 76%) and inconvenience (18/41, 44%) being the most frequent reasons.
During 44 one hour observation periods the observer recorded more HH opportunities and performances than the electronic system (mean number of HH opportunities/hour 10.9âÂ±â7.6 vs 6.8âÂ±â6.9, p <â0.001, correlation râ=â0.75, pâ<â0.001, and performances/hour 8.7âÂ±â3.9 vs 6.0âÂ±â3.1, pâ<â0.001, correlation râ=â0.60, pâ<â0.001, respectively). Correlation between observer and HH electronic system was very low (correlation coefficient râ=â0.03, p =â0.91).
The researchers concluded that the electronic HH system was not accepted by ICU staff principally due to inaccuracy and inconvenience. Inaccuracies were verified by direct observations. In order for an electronic HH system to succeed, the researchers suggest it must be highly accurate and comfortable to use.
Source: Levin PD, et al. Obstacles to the successful introduction of an electronic hand hygiene monitoring system, a cohort observational study. Antimicrobial Resistance & Infection Control. 2019;8:43