© 2023 MJH Life Sciences™ and Infection Control Today. All rights reserved.
Food and Drug Administration (FDA) investigators and U.S. Marshals have seized all implantable medical devices from Shelhigh, Inc., Union, N.J., after finding significant deficiencies in the company's manufacturing processes. The deficiencies may compromise the safety and effectiveness of the products, particularly their sterility.
The products include pediatric heart valves and conduits, surgical patches, dural patches, annuloplasty rings, and arterial grafts. The tissue-based devices are used in many surgical settings, including open heart surgery in adults, children and infants, and to repair soft tissue during neurosurgery and abdominal, pelvic and thoracic surgery. Critically ill patients, pediatric patients and immuno-compromised patients may be at greatest risk from the use of these devices.
All medical device companies must follow current good manufacturing practice, a set of requirements that help to ensure the safety and effectiveness of all medical products. According to the FDA, Shelhigh's violations include: Manufacturing products in a facility with a poorly constructed and poorly maintained clean room where sterilized devices are further processed; failing to adequately monitor critical manufacturing environments for possible microbial contamination; failing to properly test products for sterility and fever-causing contaminants; and failing to scientifically support product expiration dates.
According to the FDA, physicians should consider using alternative devices. Physicians should also monitor patients with a Shelhigh implant for infections and proper device functioning over the expected lifetime of the device. Patients who think they may have received a Shelhigh device during surgery should contact their physician for more information. FDA will issue a Preliminary Public Health Notification to physicians and other health care professionals and a Preliminary Advice for Patients shortly with more information; those documents will be posted to FDA's Web site.
The seizure follows an FDA inspection of the Shelhigh manufacturing facility last fall, as well as meetings with the company at which FDA warned Shelhigh that failure to correct its violations could result in an enforcement action. FDA also alerted the company to its manufacturing deficiencies and other violations in two warning letters.
Medical devices manufactured by Shelhigh include:
Shelhigh Pericardial Patch
Shelhigh No-React Pericardial Patch
Shelhigh No-React PneumoPledgets
Shelhigh No-React VascuPatch
Shelhigh No-React Tissue Repair Patch/UroPatch
Shelhigh Pulmonic Valve Conduit No-React Treated
Shelhigh No-React Dura Shield
Shelhigh BioRing (annuloplasty ring)
Shelhigh No-React EnCuff Patch
Shelhigh No-React Stentless Valve Conduit
Shelhigh Internal Mammary Artery
Shelhigh Gold perforated patches
Shelhigh Pre Curved Aortic Patch (Open)
Shelhigh NR2000 SemiStented aortic tricuspid valve
Shelhigh BioConduit stentless valve
Shelhigh NR900A tricuspid valve
Shelhigh MitroFast Mitral Valve Repair System
Shelhigh BioMitral tricuspid valve
Shelhigh Injectable Pulmonic Valve System